Laser Therapy Instrument to Treat Hypertension, LLLT for Diabetes, Rhinitis, Cholesterol, Cerebral Thrombosis
USB charging cable* 1
Laser Wire* 1
silicone nose plugs* 3
Power adapter* 1 (US/EU)
Chinese and English manuals * 1
Wrist laser therapeutic apparatus
A, product efficacy
1. To purify the blood
2. The flow of blood
3. To protect the brain yangxin
Second, the product scope
1. Patients with cerebrovascular disease
2. The three tenors crowd
3. The healthy people
4. People with poor sleep
5. Rhinitis crowd
Three, main technical index
Laser wavelength: 650 nm
The number of laser: six (1 nose laser + 5 wrist laser )
Single laser output power: 5 mw
Timing range: 10 to 60 minutes and six grade for adjustment
1. Lower blood sugar
2. Lower blood lipids (cholesterol, triglycerides)
3. To prevent thrombosis
4. The prevention and treatment of high blood pressure
5. Health brain, enhance memory
6. Improve the sub-health state
Regulate the body’s immune function, enhance airframe disease-resistant ability.
8. Use rechargeable batteries, can use continuously
9. The whole machine lightweight, easy to carry.
* disease of heart head blood-vessel, hyperlipemia, high blood viscosity, high blood pressure, arteriosclerosis, heart, brain, prevent thrombosis, coronary heart disease, myocardial infarction, angina pectoris, cor pulmonale, arrhythmia, cerebral thrombosis, cerebral infarction, cerebral hemorrhage and brain trauma sequela, diseases such as diabetes, vascular headache, stroke.
* respiratory diseases:
Allergic rhinitis lung infection in chronic obstructive pulmonary disease, etc
* suitable for sub-health state caused by increased blood viscosity, such as headache, dizziness, chest tightness, insomnia, tinnitus, fatigue and other symptoms.
Preparations before usage:
1. Preparations before using it for the first time
Ensure whether the usage of accessories explained well according to this instruction
There is a lithium battery in this instrument. You need to charge the battery before using it. If the battery doesn’t need to be charged any more, the screen will show you this information.
2. Precautions when charging
In order to protect the lithium battery, please use the original power adapter for Charging.
Low battery indicator
When the battery is low, this instrument will ring twice the alarm and its screen will display the information of low battery .Please turn off this instrument and change for another battery or charge the battery. (If the instrument isn’t turned off in time, it will turn off itself in 10 seconds.)
Put the laser treatment output plug into the socket as shown below.
4. Installation of therapeutic head
Before lasing, put a light guide plug on the therapeutic head as showed, and then put the therapeutic head into the nasal cavity.
Control panel operation:
1. The operation of the control panel
Starting up / Shutdown
Starting up: push the “on / off” button, it give out an opening tone.
The instrument is turned on. There is the standby interface on the display screen.
Shutdown: push the “on / off” button for three seconds any time. Timing control
Continuously press the “timing / alarm” button to set the treatment time, each push prolongs 10 minutes, and the maximum duration is 60 minutes. Repeatedly push the timing button, time displays in the pattern of circular loops.
The treatment time can be 10 – 60 minutes. A 30-minute treatment is generally recommended, 15 minutes for each nasal cavity. The timing mode is countdown. The instrument is in standby mode after giving out warning tone when time is over. If one does not continue to use it, it will automatically shut down.
2.Operation for adjusting laser power：
Choose to irradiate wrist or nasal cavity alternately by pushing “switching” button, or to irradiate both of them simultaneously.
In the case of standby interface or treatment interface, push “on-off/timing” to enter the time setting interface. It can be standby for 10 minutes, 20 minutes, 30minutes…..
Switch function: it start to work after setting the time, at this time, you can also push the “switching” button to choose to irradiate wrist or nasal cavity alternately by pushing “switching” button, or to irradiate both of them simultaneously.
Mode selection: choose different irradiation mode in the working state. There are two irradiation mode, pulse irradiation and continuous irradiation.
3.The function of automatically starting up and shutdown:
There is no keyboard or treatment for starting up, the value of laser power, low frequency intensity and timing are 0. The display screen will automatically shut down if there is no operation in 10 minutes, without influencing treatment. Then, push any button to turn on the display screen. Notes: the definition of “long push” and “short push” in the instructions: long push means more then 2 seconds; short push means less than 1 second.
Instructions for use：
Laser irradiation treatment
(1) Before the irradiation treatment, place the light nose-plug outside the laser treatment head, then put the treatment head into nasal cavity, push “switching” button or “mode” button to adjust. Patients can choose the suitable power according to habitus.
(2) Continuously push the “timing / alarm” button to set the treatment time. The treatment time can be 10 ~ 60 minutes. A 30-minute treatment is generally recommended, 15 minutes for each nasal cavity. The instrument gives out a buzzing sound when it finishes irradiation, showing the end of treatment. It automatically cut off the laser output, entering the standby state.
(3) Course of treatment: once a day, no time limit (generally, irradiation treatment in the evening is recommend), 7-10 days is one course and the interval between therapeutic courses is 5-7 days.
(4) Put the treatment head into nostril to irradiate nasal cavity, place the wire around ears to prevent
1.Avoid direct exposure to the pupil region when using laser irradiation.
2.Avoid sharing with the following electronic instruments when treating people with low frequency.
a)Implantable medical electrical equipment, such as pacemaker
b)Medical electrical equipment for life support, such as artificial heart
c)Medical electrical machines, such as Electrocardiogram Machine
3.Warning sign is on the back cap of the instrument; please take care when using it.
4.Do not attempt to open the casing; there are not any components the users can repair.
When necessary, please ask professionals for help.
Instrument can not be started
Out Of Battery Press he pow er button for 3 seconds
Recharge it immediately Please press the switch for 3 seconds for startup and shutdown
Normal display No laser output
Correct switch or wrong switch Time on or time off
Set time and then start
Laser output power decreased.
Check laser therapy head Battery low
Clean laser therapy head Recharge it immediately
Abnormal show or the button is no response
External electromagnet interference Low battery voltage
Shut down and restart the machine